# Bird Mark 7

Bird Mark 7

## Details

PRODUCT SPECIFICATIONS FEATURES • Independent flow rate control permits precise regulation of the inspiratory gas flow for flexible individualized patient care. • Simplified internal circuitry for ease of maintenance and repair. • Apneustic Flowtime simulates a dynamic (time-cycled) post-inspiratory breath hold for improved PPB therapy. (Mark 7A only) PRODUCT SPECIFICATIONS OPERATING RANGES Inspiratory Pressure . . . . . . . . . . . . . 5 to 60 cmH2O Pressure Relief (Adjustable) . . . . . . . Preset at 30 cmH2O Inspiratory Flow rate (Maximum) . . . . ≥ 68 LPM Starting Effort . . . . . . . . . . . . . . . . . . -0.5 to -5.0 cmH2O Expiratory Time. . . . . . . . . . . . . . . . . 0.5 sec to ∞ Apneustic Flowtime (Mark 7A only) . . OFF, 0.3 to 3.0 sec Apneustic Flow (Mark 7A only) . . . . . 150 - 200 ml/sec PHYSICAL WEIGHTS & DIMENSIONS DIMENSIONS H - 8.5” (22 cm) W - 10.5” (27 cm) D - 6.5” (16 cm) WEIGHT ≈ 6 lbs. (2.8 kg)

### Diverse Customer Base

• Hospitals
• SurgeryCenters
• Oncology Centers
• Education Facilities
• Home Care
• Distributors

Go to top